This device is comprised of Main Body, Handpiece, Fiber, Handpiece Connecting Line, Foot Switch, Protective Goggles, Power Supply Cord, etc. and setting on the laser output can be manipulated and operational status of the device can be confirmed through the touch screen respectively.
It is the device that applies a direct current to the laser diode by the medium of semiconductor so emits the laser of the corresponding wavelength. The handpiece is not intended to be used in contact with the skin.
808nm wavelength is of high absorbancy in hemoglobin, therefore making it appropriate for incision and coagulation of soft tissues.
Laser beam with 650nm wavelength, an additional function, deeply permeates the skin and activates functions of capillaries so it is effectively used in pain relief. Laser beam with 450nm wavelength sterilizes acne germs etc.
by inducing diffusion of endogenous (occurring naturally in the inside of skin) Porphyrin.
|Power Input and Frequency||100-240V~, 50/60Hz|
|Wavelength||405㎚, 650㎚, 808㎚|
|Output Energy / Laser safety class by IEC60825-1||a) 405㎚ : 120mW / CLASS ⅢB
b) 650㎚ : 100mW / CLASS ⅢB
c) 808㎚ : 15W / CLASS Ⅳ
|Aiming Beam||a) Medium : Laser Diode(LD)
b) Wavelength : 650nm
c) Strength : Less than 5㎽
|Type and Level of Protection against Electric Shock||Class 1, Type B|
|Optical density of safety goggle||OD6+|
|The degree of protection against harmful ingress of water||a) Main Body : IPX0
b) Foot Switch : IP68
|Laser type||Passive Cooled InGaAs Diode Laser|
|Wavelength||808nm, 650nm, 405nm|
|Laser power||15W at 808nm
100mW at 650nm
120mW at 405nm
|Output modes||Continue(CW), Pulse|
|General modes||Pulse duration 1~900ms
Pulse Repetition Rate 1~10Hz (time control 1~300sec)
|Fiber core||200um, 400um, 600um, 1000um available Fiber indication sensor|